The EC help provides an easy access to the information related to the use of Keysight EC Network Analyzer. Pressing Help key on the front panel. EB/EB, please see the Programming Manual. • Installation and Quick Start Guide (Part Number: Ex1, attached to optional ABA). This manual . The EC complies with INSTALLATION CATEGORY II as well as The Service Manual is a guide to servicing the EC ENA Series Network Analyzer .
|Published (Last):||20 August 2015|
|PDF File Size:||5.46 Mb|
|ePub File Size:||14.30 Mb|
|Price:||Free* [*Free Regsitration Required]|
This document contains proprietary information that is protected by copyright. All rights are reserved. No part of this document may be photocopied, reproduced, or translated to another language without the prior written consent of the Agilent Technologies. The printing date changes when a new edition is printed. Minor corrections and updates that are incorporated at reprint do not cause the date to change. The manual part number changes when extensive technical changes are incorporated.
Italic type is used for emphasis. If anything is missing, contact the nearest Agilent Technologies office. S5071c procedure consists of the following items. Remove the two stickers item 1. Figure 1 Item 1: Figure 3 Item 3: E Figure 4 Item 4: E Removal Procedure Step 1. Turn off the power. Inset e501c key in the lock and turn crockwise. Push the button under the manusl and remove the hard disk. Replacement Procedure Step 1.
Insert the hard disk into the slot. Turn the key to the left until it is locked. Turn on the Manyal. A dialog box appears to allow you to select a program to be loaded. RestoreSysCorFile dialog box appears Figure After restart, the screen for initial registration appears. Inset the key in the lock and turn anti-clockwise. Open the screw in clockwise direction. Press the button under the screw.
Remove the hard disk. Turn the screw in anti-clockwise direction until it is locked. Turn the key to the right until it is locked. The system recovery procedure is different in accordance with the HDD revision. The HDD revision can be checked using the following procedure: The volume label for C: If you do any operation other than the following steps, the system may not be recovered.
Types of system recoveries The following 2 types of system recoveries are available. Factory recovery Returns the contents of the C drive to the factory state. To use this function, you must prepare for recovery in advance. If the hard disk failed and has been replaced after purchase, the state when the replacement was performed is recovered.
In addition to the Windows operating system, the following settings of the EC are returned to the factory state. Network setting GPIB setting Printer setting The driver for the supported printer installed after purchase is deleted.
You need to execute initial registration again. The firmware revision you installed remains even if you perform factory recovery. Files you created using the save function files in the D drive are not affected, but we recommend backing them up before executing system recovery for precautionary purposes.
The User back up function is not available. Turn ON the EC. The boot option is displayed. Press 1, and then press Enter. A warning message is displayed. Recovery function is performed. A message box appears after the completion of recovery process. Press any key to continue. When the recovery is finished, EC restarts. CN or below This section describes how to return the contents of the C drive to the factory state.
NOTE You need a keyboard for this operation. Shut down the EC. Connect the keyboard to the EC.
Agilent Technologies E5071C Manual
Press the standby switch of the EC to turn it on. When the screen as shown in the figure below appears, press F11 of the keyboard. If you press F11 twice or more and a screen which is not mentioned in this manual appears, reboot the instrument and restart the process from the first.
After several seconds, the next screen appears automatically even if you do not press any e5017c, so do not miss it.
A confirmation dialog box is displayed. Push the enter to select Yes to start the recovery of the factory backup.
The recovery takes a few minutes depending on the amount of data. Figure Recover Factory Backup ecse Step 8. When the recovery is finished, push the enter select OK. The EC restarts automatically. Execute the calibration of the touch screen. CN or below This section describes how to create the user backup image. Close all other applications you currently open. Figure Recover User Backup ecse Step 5.
Select the C drive and D drive. Select the destination folder and input a file name, then click Next button. It is recommended to save the file to a directory which has enough space like the E drive. Also, never save the contents of the drive to the C, D or F drive. If necessary, input comments and click Next button.
The comments can be omitted. The following screen appears during backup. The following screen appears when the user backup image was created. CN or below Returns the contents of the C and D drives to a user-specified state. To use this function, you must create the user backup image in advance.
Select the saved backup image file, then click Next button. Select the C drive, then click Next button. Select the C drive to restore, then click Next button. Select the D drive to restore, then click Next button. There are two procedures in accordance with the HDD revision. Therefore you should follow the procedure which you see on the EC. You cannot use the front panel keys during the initial registration of the EC therefore connect the mouse and the keyboard before turning on the power.
If you perform the following procedure incorrectly, a message asking you whether to return to the previous registration screen and perform the registration appears. In this case, follow the instruction to return to the previous registration screen. CN or above Step 1.
Turns on the EC. In the next dialog box, input agena in the Name box. The EC restart automatically. Click OK to proceed. A progress bar appears Figure indicating the progress of the Recovery Image setup. The complete process takes less than 5 minutes e55071c finish. A message dialog box appears after the completion of the Recovery Image Setup procedure.
The dialog box disappears automatically after a few seconds. EC firmware is executed, then the measurement display appears.