A Complete HIOKI Digital & Analog Clamp Tester /A AC .. CARRYING CASE ×1. Hand strap ×1. Instruction manual ×1. Hioki Pdf User Manuals. View online or download Hioki Instruction Manual. View and Download Hioki instruction manual online. Clamp On AC/DC HiTester. Measuring Instruments pdf manual download.
|Published (Last):||19 May 2004|
|PDF File Size:||16.96 Mb|
|ePub File Size:||15.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
Hioki CLAMP ON AC/DC HiTESTER
Therefore, you must select at least one lens with this item. Our Member Store gives you access to the following advantages: Also See for Instruction manual – pages. The auto power-off function is automatically disabled.
Not finding what you’re looking for? To ask our Hioki experts a question please include your contact details. We accept these payment methods: Table of Contents Add to my manuals Add. If the function is activated when there is no input, the minimum value will remain zero. Unit turns off during use. It is recommended that a new battery should be used, when the blackout seems to be short.
Every time one minute elapses, one segment of the flashing bar graph goes on. If the unit is damaged, or fails to operate according to the specifications, contact your dealer or HIOKI representative. We have tried to bring this manual as close to perfection as we could achieve. Secondary electrical circuits connected to an AC electrical outlet through a transformer or similar device.
Refer to “Chapter4 Battery Replacement”. If perchance you find any unclear portions, mistakes, omissions, or the like, we would be most obliged if you could please notify us of them via any HIOKI agent, or directly.
Hioki 3284 CLAMP ON AC/DC HiTESTER
To print the manual completely, please, download it. Open the top ends of the clamp core and clamp the measured conductor so that it passes through the center of the clamp core. This product includes one lens. FAST Instead, the unit symbol blinks. Press to display 2. Inrush current crest value. Save my login information. Replace the battery to ensure accuracy.
A conversion table for ranges is provided below. Figs 2 and 3 As the measured value is small to the range response time becomes long.
Select a proper mode according to the waveform shown below: Next, insert the red test lead to V and the black test nioki to COM of the voltage measurement terminal. Press and select a measurement mode for the measured circuit.
Got it, manuwl to print. Yes, Sign Me Up! Continue on ITM Canada. The measurement mode is switched as follows. When all segments on the bar graph go on, the elapsed time is 30 minutes.
Remove the rear cover.
Ask an Manula More Hioki Products. Analysis for DC to distorted waves Also, when deactivating the recording function, press the key to terminate HOLD Reset all the data in a REC function. Chapter 2 Measurement Procedure Explains how to use the for measurement. You can unsubscribe at any time. If the current range of the measured circuit is known, set the current range to the manual range. We also accept wire transfers, cheques and purchase orders on approved credit.
AC/DC Clamp Meter | Clamp On AC/DC HiTester – Hioki
Remove the rear cover and insert a battery. Don’t show me this message again. Enter text from picture: